The caloric test is a part of the ENG. It is an attempt to discover the degree to which the vestibular system is responsive and also how symmetric the responses are, between left and right ears. It is a test of the lateral semicircular canals alone -- it does not assess vertical canal function or otolithic function. While not as good as we would like, the caloric test is the best that we have to deduce the function of each ear independently of the other. The rotatory chair is a better test to discover the function of the both inner ears together. The VHIT test is somewhat of a compromise -- one can often determine the bad ear, but not as well as the ENG.
Caloric testing in the USA is dying out. The reason is that doing caloric testing is generally a financial loss for the outpatient medical facility, because it takes a long time (often more than an hour), it requires an expensive piece of machinery, it requires a highly trained individual to do the test, and because many large insurance companies and Medicare pay very little for the caloric test. If you are fortunate enough to find an outpatient facility that still does good quality calorics, you are lucky !
The caloric response was first described in by Robert Barany in 1906. His findings were immediately considered pivotal and Barany received the Nobel Prize. The key observation that led to his Nobel prize was made while he was irrigating out ear wax, after a patient complained that the water was the wrong temperature. He noticed that the eyes went different directions for warm and cold water irrigation. This shows the value of being mindful.
Most caloric tests nowadays are done using a computerized system as shown below. The computer analyzes the caloric data, computing peak slow-phase velocity.
|ENG system as was used in the 1960's when we had CRT monitors (Courtesy of ICS medical). Now ENG systems are smaller. Oddly enough this illustration shows someone sitting behind the console. Practically, people are generally on an exam table with a back portion that tilts.|
There are several variant caloric tests, but at present, the standard is the water bithermal caloric test. This consists of 4 sections 2 ears * 2 temperatures (warm and cold). Ideally this is done with warm and cold water.
Calorics can also be done with air, and although not as messy, the results are not as reliable. We suggest avoiding air calorics altogether. Although air (unlike water) can be used to test a perforated ear, the accuracy is very questionable as it depends on the size of the hole in the ear drum. If the hole is large, air can get into the middle ear and produce paradoxical responses. If the hole is small, probably it is OK, but we don't think it is reasonable to take a chance that the technician doing the caloric will be able to judge whether or not the results can be trusted based on the size of the perforation. We think that the spontaneous, head-shaking, and vibration tests are a reasonable alternative in this situation.
Monothermal caloric -- 100 cc stimulus. We strongly advise against doing this, see below.
The monothermal caloric test is a variant caloric method in which a single large bolus (i.e.100 cc) of a single large bolus of ice water is given, and compared to a smaller bolus (i.e. 10 cc), rather than two irrigations with hot and cold. We advise against doing this. When mismatched irrigation volumes are used, as in the monothermal test, there is no averaging of data and thus there is lower accuracy. Another problem is that the "strong" stimulus -- 100 cc of 0 deg centegrade water (from the fridge), can provoke a gigantic response and make the patient very sick. Finally, the rationale of comparing 10 cc to 100 cc is questionable.
In the bilateral irrigation test, both ears are irrigated at the same time. This procedure is faster and results in much less dizziness than the variants where air or water are done one at a time, but it "throws away" the total response parameter, as well as any clue as to whether the stimulus reached the inner ear. For this reason, it is rarely encountered. Bilateral irrigation sacrifices too much for speed.
|Balloon version of caloric (one of the variant caloric tests, see below.) We strongly advise against doing this.|
When water is not used directly, as in the balloon test, heat or cold is not conducted as efficiently to the ear.
Because of the considerable inaccuracy even for the best of the lot, the bithermal water method, we think it is imprudent to adopt more convenient but less accurate methodology.
|Eye movement tracing during caloric in a patient with a mild bilateral reduction in responses. For the first 35 seconds, little is seen. In the next panel, a left-beating nystagmus gradually builds up. It begins to wane after roughly a minute, and at that point, fixation is attempted. This subject did not suppress the nystagmus very well.|
The caloric test is ordinarily performed with the subject reclining, head inclined 30 deg up from horizontal so as to make the lateral canal horizontal. Water is introduced into the ear canal on one side, either 7 deg centigrade above or below assumed body temperature. The flow rate is such that the ear rapidly equilibrates with the water. The water is stopped after 30 seconds, and nystagmus is observed, while the subject is distracted (usually with tasks such as naming of animals, counting backwards, etc). This is sometimes called "tasking", see following.
Nystagmus commonly builds for about 30-60 seconds, then gradually decays away over roughly 2 minutes. After a rest of at least 5 minutes, the procedure is repeated with either the opposite temperature water, or on the other side. Eye movements are usually recorded with either EOG or a video method, such as is shown on the graphic above.
Ideally subjects undergoing caloric tests should have no sedating medications for the last 24 hours. Sometimes this is difficult, as for example, when persons are addicted to medications in the Valium family. In this situation, usually 12 hours is sufficient. More data about medication effects is found here. If the individual takes medications anyway, the person doing the test should note this on the report so that the persons attempting to use this result for diagnosis will be aware that responses may be depressed.
If no response is detected, or at least none greater than the spontaneous nystagmus, then ice water should be performed. This is done with the head in the standard position on the "dead" side, and then the person is turned prone so that the head is inverted. If there is a true caloric response, the waveform will reverse. If it is just spontaneous nystagmus, the nystagmus will not be affected. A possible pitfall of this methodology is positional nystagmus.
Similarly, if there is one "outlier" response -- the operator should attempt to resolve this -- usually by repeating it.
From the peak slow-phase velocity of nystagmus four numbers are obtained -- cold right, cold left, warm right, and warm left. Spontaneous nystagmus (SN) should be subtracted from these, and then the absolute value taken. From these responses, LC, LW, RC, RW, three additional numbers are derived:
If spontaneous nystagmus is not subtracted off first, peculiar results may appear, such as greater than 100% paresis (which is obviously impossible).
There is considerable test-retest variability. The upper limits for test-retest variation for paresis were found by Proctor et al to be 24%. The upper limits for DP was 22%. This suggests that upper limits of normal should be set at greater than these values, which do not take into account variability of the normal population, and are therefore inappropriately low.
Although difficult to decode due to presentation of logarithmically transformed data, the paper by Sills, Baloh et al. 1977 has considerable useful data concerning caloric parameters.
There are many methods of distracting persons during the process of recording their nystagmus. Without distraction, responses can be suppressed which reduces validity. Tasks in which the subject produces a listing of items from memory seem reasonable and effective.
|Quiz -- i.e. "what is your age", what is your favorite color.||Less effective||Fomby et al, 1992|
|Hand-motor task, clinician directed. Touch the thumb to finger as directed by clinician.|
|Alphabet task -- third letter of alphabet following a given letter randomly selected by clinician|
|Math task, Add or subtract a number given by clinician from a running total|
|Quiz task: Name colors, states in USA, cities in ...||Best task|
|Hand-motor task -- touch thumb to first finger once, 2nd twice, third three times, etc.|
|Alphabet task #2 -- every third letter in the alphabet|
|Math task #2 -- count backwards by 3's or 7's.||Less effective|
There have been several attempts made to model the caloric response. The response is theoretically a combination, possibly nonlinear, of temperature differential induced convection stimulation of the canal, a direct effect of temperature on the nerve, transduction responses in the mechanics of the cupula, adaptation responses in the nerve and brainstem, and other central processing effects, mainly including velocity storage. A descriptive curve-fitting approach to the response is exemplified by that of Formby et al (1992, 2000).
The above is somewhat useless, as it boils down to just saying that it is very complicated. A more pragmatic way of thinking about it is to observe that the peak caloric response is largely proportional to the temperature differential across the lateral canal. The temperature differential depends on several things:
- Temperature difference between the irrigant and the inner ear (presumably body temperature)
- Thermal conductivity of irrigant (i.e. water is 10 times more conductive than air), and the part of the ear in contact with the irrigant.
- The ear drum (which has air behind it) doesn't account for much of the caloric response unless it happens to be full of fluid (usually not the case)
- Bone is the main heat conductor
- Wax is a good insulator.
- Area of bone (not TM) in contact with water. Heat is conducted mainly through the bone of the external ear canal, not the ear drum. This is proportional to the radius of the canal in contact with the irrigant.
So, simple predictions are that bigger temperature differences cause bigger calorics responses, water is more reliable than air, and that ear wax plugs should greatly decrease caloric responses.
Currently caloric testing technology does not control for most of these variables - -body temperature is not measured, there is no adjustment for air vs water in reporting norms, there is no methodology of documenting that the tip went where it went and that the ear is free of wax, and there is no adjustment for the ear canal diameter. The lack of basic controls causes variability and reduces the value of the caloric test.
While it is unusual to record the caloric response long enough, if one waits long enough (i.e. several minutes), a tiny "reversal" phase to the caloric response can be seen in normal subjects. This is generally thought to be due to adaptation processes in both the hair cells of the inner ear as well as centrally, and follows the same general train of logic of reversal phases seen after rotational stimuli. One can also elicit a similar reversal by removing gravity from affecting the lateral canals, through positioning the person back to upright. This technique has some intrinsic problems involving adding another stimlus (the tilt itself), problems with accurate positioning as well as knowledge of the canal plane in any particular individual, attempting to measure tiny amounts of nystagmus when just a small error in positioning could result in nystagmus, and potential contributions of other canals. In other words, this data is pretty hard to interpret in any rational framework.
Well anyway, Ichijo recently reported a study of 12 healthy humans and used an unusual terminology suggesting that the "second phase" was the nystagmus seen after the patient is repositioned to upright to make the lateral canals truly horizontal, and that the "third phase" was the nystagmus resulting after returning the subjects to supine (2015). We are not at all sure why one would use this nomenclature, that confuses the situation where "secondary nystagmus" is typically used for the nystagmus that occurs if you wait a long time after a stimulus (in the same position). Similar results regarding the repositioning maneuver have been reported by others (Wu et al, 2000; Aoki et al, 2006). Ichijo suggested that this secondary phase was related to perilymph pressure. A far more conventional interpretation was that of Gursel and Oosterveld (1983), who suggested that it was due to adaptation. The interested reader is referred to conventional expositions of the secondary vestibular nystagmus, such as can be found in Wilson/Melville Jones classic textbook, as well as the literature concerning caloric responses in outer space.
Fixation suppression is ordinarily evaluated by waiting till the caloric response is near peak, then allowing vision for 10 seconds, with instruction to fixate on a target. This is a close-to worthless test. The reason for this is that it is "all over the map" -- some patients fixate very well, some not at all. It depends on how nauseated they are, how well they can see without their glasses, and how cooperative they are. A more formal way of saying this is that the scatter in fixation suppression is so large, that practically any value falls within the "normal" range.
Another intrinisic problem with fixation suppression is that it is dependent on the size of the caloric response. It is much easier to suppress a 10 deg/sec response (such as due to an air caloric) than a 50 deg/sec response. To do this properly, one would need norms scaled to the caloric response. Thus the values that are produced by conventional caloric equipment have no norms. Also one would need to adjust for visual acuity (which usually is greatly reduced in persons who take off their glasses for the test). As there are so many huge flaws with the fixation suppression test, it is a result that is reported but that knowledgeable clinicians ignore.
The rotatory chair fixation test is far better controlled. It has the same problem though with visual acuity.
Acute Unilateral Vestibular Loss
|This patient has only spontaneous nystagmus on the left (about 6-8 deg/sec). On the right the traces are shifted downward due to the spontaneous nystagmus. The weakness calculation probably underestimates the amount of caloric weakness. (Figure courtesy of Dr. D. Yacovino). The most common cause of this ENG pattern is vestibular neuritis.|
Another example -- a complete loss of calorics due to vestibular nerve surgery can be found here.
|The same patient as above, one year later. Now there are clearly caloric responses on both sides, and the spontaneous nystagmus is gone . The notch in both traces at about 80 seconds shows that this patient has good fixation suppression (Figure courtesy of Dr. D. Yacovino).|
|This patient has no measurable caloric response. The most common cause of absent caloric responses is poor ENG technique (such as use of air rather than water), and after that, aminoglycoside toxicity.|
|This patient has very little caloric response. She had none at all with conventional temperatures. When ice water was used, she has a weak right-beating (similar to her spontaneous nystagmus), but it reverses to left-beating on prone. This shows that she does have a caloric response on the left.|
Completely absent caloric responses, documented with the ice-water/prone test.
This patient has NO caloric response to regular temperature. With ice water, there is a low-amplitude LB nystagmus that does not reverse direction with prone. This shows that there is a 100% vestibular loss.
The Prone test is helpful when one is attempting to determine whether or not there is any response at all. The mechanism is that caloric nystagmus will reverse prone, while spontaneous nystagmus will usually be unaffected.